Dillon XC Compression Force Gauge

Dillon XC Compression Force Gauge
currently in stock
Shipping Weight: 20 lbs. (estimate shipping cost)

Proven accuracy and ruggedness sets the Dillon XC mechanical force gauge apart in the force measurement industry. The “D” shaped deflection beam is the heart of the Force Gauge system. Highly engineered materials with close machining tolerances and proper heat treatment provide optimum strength and spring characteristics.
Ideal for stand alone force applications or often used as the load defining component of a much larger system.
The Dillon XC compression calibration force gauge comes in 11 capacities ranging from 25 lb to 50,000 lb or 10 to 20000 kg. All XC force gauges feature accuracy of ±1% of full capacity, except the 25,000 and 50,000 (10,000 & 20,000 kg) capacity instruments which are accurate to ±2% of full capacity.
Load is applied against a hardened ball which rotates to maintain vertical alignment as pressure increases. The ball is held in place with a spring clip or retainer. A threaded mounting hole is located opposite the loading ball in the bottom of the beam.
Optional shockless dial indicator is available for installations involving the sudden application or release of force (maximum pointer cannot be supplied with shockless dial indicator). A maximum load pointer is also available, which remains at peak load until manually reset.
The Dillon XC compression gauge comes with force gauge assembly, carrying case and compressionloading spherical ball fitting (note: case is not included with the 25000 lb, 50000 lb, 10000 kg or 20000 kg models).
Standard Compression Pounds Models
Model 
303860035 
303860043 
304460033 
304460017 
304440019 
303880017 
303890016 
304490014 








Price 
$743.75

$743.75

$743.75

$756.50

$977.50

$977.50

$1215.50

$2516.00

Capacity & Dial Divisions 
50 x 1/2 lb 
100 x 1 lb 
250 x 21/2 lb 
500 x 5 lb 
1000 x 10 lb 
2000 x 20 lb 
5000 x 50 lb 
25000 x 250 lb 
Compression Models with Maximum Hand in Pounds
Model with Maximum Hand 
303860159 
304460090 
304460074 
304440050 
303880058 
303890057 
304230055 
304490055 








Price 
$816.00

$816.00

$816.00

$1045.50

$1045.50

$1287.75

$1959.25

$2601.00

Capacity & Dial Divisions 
100 x 1 lb 
250 x 21/2 lb 
500 x 5 lb 
1000 x 10 lb 
2000 x 20 lb 
5000 x 50 lb 
10000 x 100 lb 
25000 x 250 lb 
Compression Models with Maximum Hand in Kilogram
Model with Maximum Hand 
303860183 
304460181 
304460082 
304440068 
303880066 
303890065 
304230063 
304490063 








Price 
$816.00

$816.00

$816.00

$1045.50

$1045.50

$1287.75

$1959.25

$2601.00

Capacity & Dial Divisions 
50 x 0.5 kg 
100 x 1 kg 
200 x 2 kg 
500 x 5 kg 
1000 x 10 kg 
2000 x 20 kg 
5000 x 50 kg 
10000 x 100 kg 
iDillon XST Force Gauge with Tension Calibration

Dillon XST Tension Force Gauge
currently in stock
Shipping Weight: 15 lbs. (estimate shipping cost)

Proven accuracy and ruggedness sets the Dillon XST mechanical force gauge apart in the force measurement industry. The “D” shaped deflection beam is the heart of the Force Gauge system. Highly engineered materials with close machining tolerances and proper heat treatment provide optimum strength and spring characteristics.
Ideal for stand alone force applications or often used as the load defining component of a much larger system.
The Dillon XST tension calibration force gauge comes in 9 capacities ranging from 25 lb to 10,000 lb or 10 to 5000 kg. Accuracy is ±1% of full range (note: for applications requiring capacities beyond 10,000 lb or 5000 kg in tension, consider the Dillon dynamometers).
Dillon XST tension force gauges in capacities through 2000 lb (1000 kg) are supplied with two rod end connectors. 5000 lb, 10000 lb, 2000 kg and 5000 kg capacities are equipped with convenient shackles and pins.
The Dillon XST tension gauge comes with force gauge assembly, carrying case and tension hardware.
Standard Tension Pounds & Kilogram Models
Model 
304430044 
304450034 
304450018 
302760012 
304400013 

304430093 
304450026 








Price 
$752.25

$752.25

$769.25

$990.25

$990.25


$752.25

$769.25

Capacity & Dial Divisions 
100 x 1 lb 
250 x 21/2 lb 
500 x 5 lb 
1000 x 10 lb 
2000 x 20 lb 

50 x 1/2 kg 
200 x 2 kg 
Tension Models with Maximum Hand in Pounds
Model with Maximum Hand 
304430150 
304450109 
304450083 
302760053 
304400054 
304420052 
304410053 







Price 
$824.50

$824.50

$841.50

$1058.25

$1058.25

$1364.25

$2426.75

Capacity & Dial Divisions 
100 x 1 lb 
250 x 21/2 lb 
500 x 5 lb 
1000 x 10 lb 
2000 x 20 lb 
5000 x 50 lb 
10000 x 100 lb 
Tension Models with Maximum Hand in Kilograms
Model with Maximum Hand 
304430176 
304430184 
304450182 
304450091 
302760061 
304400062 
304410061 







Price 
$824.50

$824.50

$824.50

$841.50

$1058.25

$1058.25

$2354.50

Capacity & Dial Divisions 
25 x 1/4 kg 
50 x 1/2 kg 
100 x 1 kg 
200 x 2 kg 
500 x 5 kg 
1000 x 10 kg 
5000 x 50 kg 
Dillon XPP Tension / Compression Calibration Push Pull Gauge

Dillon XPP Tension / Compression Calibration Push Pull Gauge
currently in stock
Shipping Weight: 15 lbs. (estimate shipping cost)

Proven accuracy and ruggedness sets the Dillon XPP mechanical force gauge apart in the force measurement industry. The “D” shaped deflection beam is the heart of the Force Gauge system. Highly engineered materials with close machining tolerances and proper heat treatment provide optimum strength and spring characteristics.
Ideal for stand alone force applications or often used as the load defining component of a much larger system
The Dillon XPP compression / tension calibration push pull gauge comes in 7 capacities in pounds ranging from 50050 lb up to and including 500005000 lb and 4 metric capacities from 50050 to 100001000 kg. Accuracy is ±1% of maximum dial reading (based on total capacity of both compression and tension scales).
Dillon Xpp gauges in capacities up to and including 100001000 lb or 5000500 kg are supplied with a set of selfaligning spherical rod end connectors for tension loading. Force is applied to connectors through a hardened steel pin which must be slip fit in connector holes.
250002500 and 500005000 lb and 100001000 kg capacity gauges are equipped with two shackle adapters, shackles and pins. Shackles must be removed when compression load is involved. Force is then applied against shackle pins in a suitable test setup.
The Dillon XST tension gauge comes with force gauge assembly, carrying case, compression loading spherical ball fitting for compression loading and tension hardware.
Pound Versions 
Model 
Capacity & Dial Divisions 
Price 
307950014 
50050 x 1 
$901.00 
307960013 
1250125 x 2.5 lb 
$901.00 
307980011 
5000500 x 10 lb 
$1041.25/div> 
308000017 
250002500 x 50 lb 
$1564.00 

Kilogram Versions 
Model 
Capacity & Dial Divisions 
Price 
307960021 
50050 x 1 kg 
$901.00 
307980029 
2500250 x 5 kg 
$1041.25 

Kilogram Versions 
Model 
Capacity & Dial Divisions 
Price 
307960021 
50050 x 1 kg 
$901.00 
307980029 
2500250 x 5 kg 
$1041.25 
308000025 
100001000 x 20 kg 
$1564.00 

Dillon XSeries Force Gauge Parts Identification
Principle of Operation
A Dshaped deflection beam is the heart of the Dillon X force gauge. Machined to close tolerances, beams are heat treated to develop optimum strength and spring characteristics. Highstrength aluminum is used in Dillon X series instruments through 200 lbs (200 kg). Ranges above this are fabricated from aircraftquality alloy steel.
A precision dial indicator is mounted at the null point of the deflection beam. The indicator plunger rests against a slanted anvil at the open end of the beam, as shown in the photo. Under compression loads, the two halves of the beam tend to close. Tension force causes them to move apart. This action pushes the plunger inward, as determined by the slant of the anvil. Readings produced on the dial are in direct relation to applied load. The pointer revolves 360 degrees clockwise under compression or tension forces. Pushpull gauges read half scale (180 degrees) clockwise in compression, counterclockwise, 180 degrees from center zero under tension loads.
Low Beam Deflection
When measured across the center of top and bottomloading holes, the approximate beam deflection is as follows:
 25 to 250 lb (10 to 100 kg) capacity  0.019"
 500 lb (200 kg) capacity  0.016"
 1000 lb (500 kg) capacity  0.018"
 2000 lb (1000 kg) capacity  0.011"
 5000 & 10000 lb capacity (2000 & 5000 kg)  0.010"
 25000 & 50000 lb capacity (10000 & 20000 kg)  0.022"
Overload Rating
Accidental overloads of 30% of capacity can be safely sustained without injury to the dial indicator or deflection beam. All capacities feature a 5:1 safety factor.
General Information
 To reset zero, loosen knurled bezellocking screw and rotate dial
 Dillon X series force gauges can be mounted horizontally, vertically or flat
 The baked enamel finish resists corrosion and rust
 Operating temperature up to 120 degrees fahrenheit
Dimensions
Model XC Compression Gauges

Pounds 
Kilograms 
A 
B 
C 
D 
E 
F 
G 
H 
J 
J1 
K 
A 
25 x 0.25 
10 x 0.1 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
50 x 0.5 
25 x 0.25 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
100 x 1 
50 x 0.5 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
250 x 2.5 
100 x 1 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
500 x 5 
200 x 2 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
1000 x 10 
500 x 5 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/428" 
1/428" 
2.94" 
2000 x 20 
1000 x 10 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/220" 
1/220" 
3.44" 
5000 x 50 
2000 x 20 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2" 
2.25" 
0.38" 
1/220" 
1/220" 
3.44" 
B 
10000 x 100 
5000 x 50 
5.87" 
1.87" 
0.94" 
3.94" 
2.75" 
3" 
3.62" 
0.75" 
7/814" 
7/814" 
4.5" 
25000 x 250 
10000 x 100 
6.56" 
2.38" 
1.18" 
4.68" 
3.31" 
3.63" 
3.62" 
1" 
114" 
11/212" 
5.5" 
50000 x 500 
20000 x 200 
7.59" 
2.75" 
1.38" 
5.5" 
3.28" 
4" 
3.62" 
1" 
114" 
13/412" 
6.31" 
Model XST (Tension) and XPP (PushPull) Gauges
Pounds 
Kilograms 
A 
B 
C 
D 
E 
F 
G 
H 
J 
K 
L 
M 
N 
 
 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.25" 
0.75" 
0.38" 
5" 
4.25" 
1/428 
 
 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.25" 
0.75" 
0.38" 
5" 
4.25" 
1/428 
50050 lb 
 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.25" 
0.75" 
0.38" 
5" 
4.25" 
1/428 
1250125 lb 
50050 kg 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.25" 
0.75" 
0.38" 
5" 
4.25" 
1/428 
2500250 lb 
1250125 kg 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.25" 
0.75" 
0.38" 
5" 
4.25" 
1/428 
5000500 lb 
2500250 kg 
4.25" 
1" 
0.5" 
2.5" 
2.06" 
2.03" 
2.25" 
0.5" 
1.31" 
0.62" 
6.94" 
5.62" 
1/220 
100001000 lb 
 
4.25" 
1" 
0.5" 
3" 
2.25" 
2.19" 
2.75" 
0.5" 
1.31" 
0.62" 
7.44" 
6.12" 
1/220 
250002500 lb 
100001000 kg 
4.25" 
1" 
0.5" 
3" 
2.25" 
2.66" 
2.75" 
10.94" 
1.69" 
0.75" 
5.44" 
1.94" 
1/220 
500005000 lb 
 
5.88" 
1.88" 
0.94" 
3.94" 
2.75" 
3" 
3.62" 
12.69" 
1.69" 
0.75" 
71.9" 
1.94" 
7/81/4 

Tension (lb) 
Tension (kg) 
C 
25 x 0.25 lb 
10 x 0.1 kg 
50 x 0.5 lb 
25 x 0.25 kg 
100 x 1 lb 
50 x 0.5 kg 
250 x 2.5 lb 
100 x 1 kg 
500 x 5 lb 
200 x 2 kg 
1000 x 10 lb 
500 x 5 kg 
D 
2000 x 20 lb 
1000 x 10 kg 
5000 x 50 lb 
2000 x 20 kg 
10000 x 100 lb 
5000 x 50 kg 